Advanced biosensors for detection of pathogens related to livestock and poultry
© The Author(s) 2017
Received: 28 October 2016
Accepted: 23 January 2017
Published: 21 February 2017
Infectious animal diseases caused by pathogenic microorganisms such as bacteria and viruses threaten the health and well-being of wildlife, livestock, and human populations, limit productivity and increase significantly economic losses to each sector. The pathogen detection is an important step for the diagnostics, successful treatment of animal infection diseases and control management in farms and field conditions. Current techniques employed to diagnose pathogens in livestock and poultry include classical plate-based methods and conventional biochemical methods as enzyme-linked immunosorbent assays (ELISA). These methods are time-consuming and frequently incapable to distinguish between low and highly pathogenic strains. Molecular techniques such as polymerase chain reaction (PCR) and real time PCR (RT-PCR) have also been proposed to be used to diagnose and identify relevant infectious disease in animals. However these DNA-based methodologies need isolated genetic materials and sophisticated instruments, being not suitable for in field analysis. Consequently, there is strong interest for developing new swift point-of-care biosensing systems for early detection of animal diseases with high sensitivity and specificity. In this review, we provide an overview of the innovative biosensing systems that can be applied for livestock pathogen detection. Different sensing strategies based on DNA receptors, glycan, aptamers and antibodies are presented. Besides devices still at development level some are validated according to standards of the World Organization for Animal Health and are commercially available. Especially, paper-based platforms proposed as an affordable, rapid and easy to perform sensing systems for implementation in field condition are included in this review.
Infectious diseases are the leading causes of death of humans and animals worldwide. Wildlife and domestic animals pathogen infections threat animal production and food supply, seriously impact animal welfare and have potential environmental and global biodiversity consequences. There is a clear economic cost of animal infectious disease as they impact large-scale developmental projects. In addition, viral infections of animal population carry global public health risks of sporadic human zoonotic infections or emergence of a pandemic viral strain. Animals are thought to be the source of more than 70% of all emerging infections .
One of essential elements for implementation of an efficient response to infectious disease threats is a rapid, selective and sensitive assay for pathogen diagnostics. Current research attempts to adopt a multidisciplinary approach for both identification of underlying pathogenic agents and control infectious diseases spread. Certainly, early detection of pathogen is crucial for managing infections and establishing improved decision-making tools.
Conventional methods for viral detection include virus or microorganism propagation and isolation from culture. These methods are effective and sensitive but tend to be costly, labor intensive and time consuming (typically results are available in 2–10 days). Alternative molecular methods based on polymerase chain reaction (PCR), real time PCR (RT-PCR) are more specific, sensitive and take less time, but they need isolated genetic materials, manipulation with special care and necessitate sophisticated equipment, and, thus, they are hardly to be applied for on-site monitoring. Consequently, development of a valid diagnostic assay for swift pathogen detection and identification, with high sensitivity and selectivity is a challenge for researchers all over the world.
Biosensors, as analytical devices, are attractive solutions for fast and efficient infectious disease diagnostics due to their simplicity, possible miniaturization and potential ability for real-time analysis [2–6]. Over the past 30 years, a number of biotechnological innovations have provided biosensors for bacterial and viral detection and monitoring. Some emerging systems have resulted in promising prototypes that achieved rapid pathogen detection without demanding high level of sample manipulation which is highly inconvenient for infected samples.
In this review we will focus on biosensors that can be applied for domestic animal pathogen diagnostics. Different biomarkers of animal infectious diseases (as proteins, DNA, RNA) and commonly used in biosensing technologies, especially for virus detection are considered in details. Examples are given for pathogens responsible for major economic losses in cattle, pig, sheep and poultry farming.
2 Principal of biosensing technology
There are two principal challenges to develop a biosensor for pathogen detection: (i) elaboration of a bioassay for biomarker detection, and (ii) improving the robustness of the bioassay to adapt it for applications in field and/or on complex biological samples. Indeed, many bioassays that work well on the bench with purified biomarker molecules fail to detect them in complex media like blood or serum. In addition, diagnostics of infection disease require high sensitivity since pathogens might spread rapidly before that any clinical sign appears in animals.
3 Detection of Escherichia coli
Escherichia coli (E. coli) is a Gram-negative rod-shaped bacterium, diversified into harmless strains, normally found in the lower intestine microbiota of humans and animals, and virulent strains that cause infections, including gastroenteritis, urinary tract infection, meningitis, peritonitis, and septicemia. In poultry, E. coli causes colibacillosis characterized by the migration of the virulent strains, as O78:K80, O1:K1, and O2:K1, from intestine to other organs as respiratory or urinary tracts. These infections decreases egg production, reduce chicken grow, increase mortality and cause important economic losses. For instance, in 2011, an outbreak of a highly virulent E. coli in Europe resulted in subsequent food withdrawals from the market and export bans leading to about $417 million negative economic impact for EU farmers . Colibacillosis is also seen in a variety of farm animals like cattle, pigs, goats  and has a significant economic importance concerning the loss of livestock. In cattle, pathogenic variants of E. coli are responsible for diseases, as septicemia and diarrhea in newborn calves or acute mastitis in dairy cows. The use of antibiotic in colibacillosis treatment and prevention is become an even greater problem than the infection itself. The extensive antibiotic use in animal production has incidence on the spread of multidrug-resistant bacteria and on antibiotic-resistant infections in humans. In most developed countries, serious consideration is being undertaken to regulate and severely restrict the use of antibiotics in animal production. The rapid and accurate diagnostic of virulent E. coli strains is vital for assessing the antibiotic resistance information, for administration of appropriate treatment and thus for avoiding useless antibiotics utilization.
The main issue in E. coli diagnostic is to distinguish between the closely related strains in order to distinguish between pathogenic and non-pathogenic species. Commonly used methods for E. coli detection and identification include culture, fermentation, enzyme linked immunosorbent and PCR assay. These methods show disadvantages in terms of long identification time (typically few days), high labor and reagent cost [9, 10]. Novel biosensors based on specific biochemical recognition strategies have been reported for rapid and specific E. coli detection as those based on PCR , quartz crystal microbalance system , surface plasmon resonance , chemiluminescence  and electrochemistry [15–17]. The biosensors for E. coli diagnostics are elaborated to assure two major steps: the capture of target bacteria from the biological or environmental samples and subsequent identification of captured bacterial sub-type. A huge variety of high affinity antibodies against E. coli that bind to surface/flagella proteins is available, as well as appropriate labels that can be employed for the amplification of detectable signal (as enzymes, biofunctionalized nanoparticles or fluorophores). These allow development of various sandwich-type immunosensors and immunoassays for rapid detection of E. coli in infected samples.
Immunosensor developed by Jaffrezic-Renault et al.  is based on the following strategy: addressable magnetic nanoparticles coupled with anti-LPS antibodies were used for the generic capture of Gram-negative bacteria onto the graphite ink electrode. The use of immunomagnetic beads allow detection of a biomarker contained in complex sample matrices. Conductometric measurements allowed real-time, sensitive detection of E. coli or Serratia marcescens cultures from 1 to 103 CFU/mL. The conductometric immunosensor permitted also the direct detection of 10 to 103 CFU/mL of Pseudomonas aeruginosa and Acinetobacter baumannii strains that were undetectable using standard immunoblot methods. Gram-positive bacteria such as Staphylococcus epidermidis were not detected indicating the specificity of detection .
Eltzov and Marks [20, 21] have proposed a point-of-care detection system based on stacked paper membranes that quantify E. coli within less than 5 min. When liquid samples containing bacteria are added onto the bottom membrane layer, the liquid starts to migrate from the lower to the upper layers. As each layer becomes wet, E. coli cells from contaminated samples push through to the next membrane layers. During migration, the bacterial cells bind with the specific antibody, itself conjugated with horseradish peroxidase (HRP) enzyme to allow signal measurement. In target-free samples, the HRP-labeled anti-E. coli antibodies from migrating are stopped by previously immobilized target bacteria on the capture layer. The upper-most layer contains the HRP enzymatic substrate producing a measurable signal only with samples containing target E. coli. This portable and rapid immunoassay was shown to have around 1000-folds higher sensitivity than ELISA since only 100 cells/mL were successfully quantified.
4 Detection of avian influenza viruses
Aquatic birds constitute the main reservoir for avian influenza viruses (AIVs) . These viruses represent a global threat to animal health and international poultry industry. Particularly high concern represents pandemic emergences which may cause enormous economic losses. AIVs are divided into low and highly pathogenic strains regarding their pathogenicity for chicken. The highly pathogenic AIV (HPAIV) spreads rapidly in domestic poultry and result in high mortality rate. The low pathogenic AIV (LPAIV) may cause mild respiratory or gastrointestinal symptoms, but usually without any signs of illness. However, LPAIV strains may acquire high pathogenicity during multiple infections in a chicken population .
Based on their antigenic specificity, Influenza A viruses infecting birds, are divided into 16 hemagglutinins (HA, H1-16) and 9 neuraminidases (NA, N1-9) subtypes. To date, naturally occurred HPAIV that produce high mortality in chickens, turkeys and other birds of economic importance have been restricted only to H5, H7 and H9 subtypes. Of these viruses, some H5N1, H7N2 and H7N7 viruses have shown to cause up to 100% mortality with 48 h in infected chickens. Many outbreaks of HPAIV occurred in domestic poultry production systems as it was the case with the highly pathogenic avian influenza H5N8 virus since 2014 . H5N1 viruses have affected the poultry industry in numerous countries for the past 15 years and have resulted in the deaths of millions of birds. Consequently, a global influenza virological surveillance in poultry populations and migratory birds is recommended by both by World Organization for Animal Health (OIE) and World Health Organization (WHO) [25, 26]. Availability of a swift and sensitive AIV diagnosis tool that allows virus strain identification will be highly useful for disease surveillance as well as for optimizing biosecurity measures on farms.
Avian influenza virus belong to the Influenza A virus family Orthomyxoviridae. The genome of AIV is constituted by eight segments of negative-stranded RNA. Ten main proteins are encoded by viral genome: the polymerase basic protein 1 (PB1) and 2 (PB2), hemagglutinin (HA), nucleoprotein (NP), neuraminidase (NA), the polymerase acidic protein (PA), matrix proteins 1 and 2, and non-structural protein 1 and 2 . Some other influenza A virus proteins were found as PB1-F2, PB1-N40, PA-X, NEP, M42, PA-N155, PA-N182 [28–31]. Potentially all these viral proteins, as well as their corresponding coding RNA sequences, are AIV biomarkers. Nevertheless, frequent mutations in the AIV genome lead to changes in antigenic properties which restrained the choice of biomarkers. HA, M2 and NA are the dominant targets for the host antibody response, anti-viral drug development and represent also the targets in diagnostics assays.
A broad of traditional serological diagnostic methods like haemagglutination (HA) test, Hemagglutination-inhibition (HI) test, neutralization (NT) and ELISA test are available as well as those based on virus propagation and isolation from cell culture or embryonated chicken eggs. These methods are effective and sensitive but they require relatively important amount of virus particles and special sample collection and handling tend to be costly, labor intensive and time-consuming. Moreover, HPAIV, as H5N1, are quite virulent for eggs, killing them quickly. This makes standard egg culture amplification procedure quite difficult. Alternative molecular methods based on PCR and RT-PCR are more sensitive but need extracted genetic material and require the use of equipment which is available only in diagnostic or scientific laboratories. In consequence, on-site detection of avian influenza viruses is rare until now for both early diagnostics and monitoring [32–34]. Similarly, a variety of different serological tests that detect the response of the infected host are established but they are not adapted to strains which have a pandemic potential. Moreover they are not robust and are difficult to be employed in point-of-care condition . There are several bedside tests which allow a relatively quick (up to 30 min) detection of viral antigens . Unfortunately, these tests provide low sensitivity and often produce false negative results, especially during later stages of the disease development. Low sensitivity is the main reason why these tests are rarely used in routine diagnostics of influenza virus.
Samples to be detected for influenza virus by lateral flow test are solubilized in a detergent containing solution and deposited on the sample pad. This step initiates lateral flow of sample components. In the first step of detection a specific influenza virus biomarker is recognized by antibodies carried by gold-nanoparticles pre-adsorbed on the conjugate pad. Usually antibodies raised against preserved epitopes in viral nucleoprotein are used for this step. Influenza viruses complexed with immune-gold nanoparticles reach the test lines. Two test lines with pre-immobilized antibodies that specifically recognize either influenza A or B virus bind to different epitopes of the virus and, in that way accumulate immune-gold nanoparticles carrying the viruses. The accumulation of gold nanoparticles results in an appearance of a visible red line. Usually, antibodies recognizing specific epitopes in HA proteins are used for the test line. Finally, non-bound immune-gold nanoparticles arrive at the control line which harbors the secondary antibody, showing the second visible red line. In the absence of viral particles in the sample, the immune-gold nanoparticles flow alone and bind only to the control line. Thus, two colored lines stand for positive result while a single colored line corresponds to negative result (Figure 3). In most cases antibodies can distinguish between influenza A and B viruses but still not able to differentiate within subtypes and, thus, between LPAIV and HPAIV stains. In contrast, aptamers generated against specific AIVs or specific RNA/DNA primers immobilized on the test lines allow AIV sub-typing.
The extreme simplicity to use, efficiency, reliability and label-free detection characterized lateral flow tests. However, their sensitivity is sufficient for detecting proteins but has to be improved to allow detection of viral particles in a complex medium as fecal swab sample. It was shown that silver nanoparticles added to the test line can amplify the colorimetric signal up to 1000-fold . A limit of detection 0.09 ng/mL was estimated for AIV detection by lateral flow test amplified with quantum dots . For comparison, this test showed 100-fold higher sensitivity than ELISA performed with the same antibodies. Recently, a quantum-dot based lateral-flow immunoassay system was proposed for quantitative detection of influenza A virus subtypes H5 and H9 . The modification of specific antibodies with quantum-dot amplified signal and permitted a quantitative read-out of the virus detection under an ultraviolet lamp.
Recently, an electrochemical immunosensor based on a specific anti-M1 antibody was shown to detect all serotypes of influenza A virus  with sensitivity similar to classical molecular methods (80–100 × 103 PFU/mL). A lower effective limit of 1 × 103 PFU/mL was achieved by coupling the anti-M1 monoclonal antibody to gold nanoparticles in a quartz crystal microbalance assay . A sensitive plasmon-assisted fluoro-immunoassay was developed for the detection of the influenza virus by specific anti-M1 antibodies conjugated to gold nanoparticle-decorated carbon nanotubes. After influenza virus binding to these mixed nanoparticles, a fluorescent signal was produced by addition of cadmium telluride quantum dots. A photoluminescence intensity of quantum dots was shown to vary as a function of virus concentration, with a detection limit of 50 PFU/mL. In another study, it was demonstrated that an electrochemical immunosensor provides a very sensitive platform for detection and quantification of PB1-F2 protein of Influenza A virus in infected cells [51, 52]. The detection limit of the device was determined as 0.42 nM PB1-F2 . A PCR-free paired surface plasma wave biosensor has been successfully developed for detection of 2009 pandemic influenza A virus . The proposed diagnostic method is rapid, sensitive and accurate but only applicable for laboratory diagnostics.
Many groups develop multiplex detection of different influenza subtypes on a single device. Microarrays for diagnostic applications are elaborated either for monitoring pathogen virulence genes or for simultaneous screening of multiple pathogens. The majority of influenza microarray utilizes a panel of primers for multiplex PCR amplification of the HA, NA and M1 genes of influenza viruses. Li et al.  coupled a DNA microarray and multiplex reverse transcriptase PCR microarrays to provide subtyping of influenza virus strains. Similarly, Kessler et al.  developed a three-dimensional DNA flow-through biochip for typing and subtyping of influenza viruses. Wang et al.  designed a microarray-based detection and genotyping of a wide range of respiratory viruses. In one another study, Townsend et al.  described a microarray, for the relatively rapid identification of influenza A virus subtypes H1N1, H3N2 viruses circulating in the human population as well as the highly pathogenic avian A/H5N1 virus that was isolated from poultry in Southeast Asia and that spread to Europe. This test demanded extraction and amplification of the viral RNA, and thus cannot easily be used on-site. However, the device provided the absolutely correct types and subtypes for an average of 72% of the isolates. The authors emphasized that the failures were due to the nucleic acid amplification step rather than limitations in the microarray.
Although microarrays are typically used a fluorescent read-out there is a great potential of electrochemical microarrays for detection of influenza viruses. For instance, the microarray silicon chip for the detection of influenza A virus based on more than 12 500 electrodes, each carrying specific oligomers allowed the genotyping and identification of HA and NA influenza A proteins and virus subtyping . Current developments are focused on miniaturization and automatization of microarray biosensors for building portable diagnostics platforms for avian influenza viruses.
5 Detection of Mycoplasma and other bovine mastitis pathogens
Mycoplasmas are the smallest bacterial cells that lack a cell wall around their cell membrane, which makes them insensible to many common antibiotics. Antibiotics help preventing some clinical signs but cannot eliminate infection. Various mycoplasmae strains infect animals, but usually play a secondary role in infections by exacerbating pre-existing disease. Nevertheless, Mycoplasma bovis (M. bovis) can play a primary role of pneumonia, mastitis or arthritis in cattle. M. bovis is considered as one of the most pathogenic and the most frequent Mycoplasma species. It is estimated that M. bovis infection causes €144 million in the European cattle industry . Methods used for diagnosis of Mycoplasma infection include bacterial culture, fluorescent antibody-based test, serological tests and PCR. Serological tests cannot be applied for early diagnosis or for detection of acute infection since serum antibody biomarkers rise at 10–14 days after acute infection. The somatic cell count as a reference method for monitoring milk quality allows mastitis diagnosis. This simple but labor-intensive assay implies methods as microscopic analysis or cytometry for raw milk. Both methods are slow and require well-trained staff to provide result accuracy and reproducibility. Early diagnosis is however of the extreme importance due to the high costs of treatments. Some portable somatic cell counters have been validated and are used as cow-side test for mastitis control at a dairy farm. Although they are very simple to apply, they have low sensitivity and specificity and usually cannot identify the pathogen strain [65, 66]. Especially, the early diagnosis of bovine mastitis is important regarding its huge impact on farm economics due to treatment cost and reduction in milk production.
Biosensors have been developed to detect specific Mycoplasma biomarkers in a rapid and easy-to-use diagnosis manner. Majority of in-field biosensors are developed to detect NAGase and haptoglobin biomarkers in milk samples. Expression of both bacterial proteins is characteristic for the acute phase inflammation. For instance, oxido/reduction process of 1-naphthol, which is a substrate of NAGase can be easily detected and quantified using carbon electrodes. Pamberton et al.  have reported an electrochemical sensor based on a screen-printed carbon electrode which detected NAGase protein in milk samples with a limit of detection of 10 mU/mL. Surface plasmon resonance was employed to monitor binding of haemoglobin to haptoglobin immobilized on the surface of the chip . The formation of this protein complex resulted in changes in mass attached to the chip surface which indicates mastitis. Though, when this biomarker is targeted to test milk samples false positive results are obtained in some milk samples that contained blood traces.
A single-stranded DNA aptamer showing high affinity and specificity against P48 protein of M. bovis has been used in a competitive enzyme-linked aptamer assay for the detection of M. bovis in sera . P48 protein is an optimal biomarker for M. bovis since it is an invariable protein that is localized on the membrane surface of M. bovis. A competitive enzyme-linked aptamer assay using the biotin-labeled aptamer of P48 protein was applied in an indirect diagnostic test. The sensitivity and selectivity of the test were similar to commercial ELISA kits.
Mastitis as an inflammatory infection may be caused not only by Mycoplasma, but by many other bacteria. The most frequent clinical infections in dairy cattle are caused by about ten different bacterial pathogens as Staphylococcus aureus, Streptococcus agalactiae, S. bovis, S. canis, E. coli. Consequently for controlling the disease spread and for targeting antimicrobial therapy a rapid strain identification and quantification of the microbial load in milk samples is needed. Nowadays, PCR or RT- PCR-based tests are employed for laboratory testing. As mentioned above, these methods are rapid and sensitive but their application for milk samples is sometimes impeded as milk contains calcium ions and proteinases that act as PCR inhibitors. Various procedures are established for DNA extraction and purification from the raw milk. Reagents for these additional steps are included in commercial kits which additionally increase reaction time and the price per analysis. Nevertheless some PCR-based kits allow identification of total bacterial strains causing mastitis and detection of their antibiotic resistant gens .
An electrochemical sensor based on electrochemical impedance spectroscopy was developed for detection of pathogenic Staphylococcus aureus ATCC25923 bacteria by Braiek et al. . A linear relationship between the increment in the electron transfer resistance and the logarithmic value of S. aureus concentration was observed between 10 and 106 CFU/mL. The limit of detection was low as 10 CFU/mL, and the reproducibility was calculated to 8%. In addition, a good selectivity versus E. coli and S. epidermidis was demonstrated.
6 Detection of Clostridium perfringens
Clostridium perfringens (C. perfringens) is a gram-positive, anaerobic, fermentative, spore-forming soil bacterium that may produce one or several exotoxins. At least 17 different C. perfringens toxins have been identified [75, 76], some of them being very potent causing animal and food-borne illnesses, and are even considered to be bioweapons. C. perfringens are classified into five isotypes (A, B, C, D and E) regarding five major toxins they produce [alpha, iota, alpha, beta (1 and 2), epsilon toxins, and enterotoxin E] . In addition, novel toxins have been identified, as NetB, which was isolated from chickens with necrotic enteritis . The toxin productions in animal gut are associated with specific enterotoxemias characterized by a variety of symptoms and traumatic infections. C. perfringens strains may infect birds, dogs, horses, pigs, lambs, cattle, sheep, goats and other domestic animals. Although C. perfringens is an important enteric pathogen for almost all domestic animals, the most advance studies on its pathogenicity have been made for broiler chicken. For instance, alpha- and beta-toxin of C. perfringens are responsible for clostridial enteric disease in poultry. This necrotic enteritis cause significant economic losses to the global poultry industry due to high mortality rates. Furthermore, some strains of C. perfringens that produce enterotoxins at sporulation are responsible for foodborne disease in humans.
The isolation of C. perfringens alone is not sufficient for diagnosis but has to be confirmed with histological evaluation of lesions. The lesions in the lamina propria associate with a strong inflammation is characteristic for early stages of infection, while in later stages, necrosis of mucosa and association of gram-positive rods with the lesions is characteristics . The infections are usually treated by antibiotics. In farms that stopped using antibiotic growth promoters, outbreaks of clostridial infections increase dramatically. In consequence, the improving diagnostics and surveillance of C. perfringens are crucial.
Diagnosis of enteric diseases produced by C. perfringens is usually difficult since these bacteria can normally colonize bird gut. Moreover, it was shown that alpha-toxin of C. perfringens is not an essential virulence factor in the pathogenesis of necrotic enteritis in chickens. Thus, the main challenge is to determine the biological activity of C. perfringens in terms of rapid strain identification and differentiation of pathogenic from non-pathogenic strains. A solution can be found in cell-based detection systems which are emerging biosensor technologies that detect the biological activity of pathogens or toxins. Those biosensors have mammalian cells as sensing elements and allow monitoring perturbations in cell physiological activities following exposure . Cell-based biosensors are capable of detecting the presence of pathogens or active toxin in clinical, environmental and food samples  rendering accurate estimation of the risk associated with the agent identification. Different device designs are proposed from a 96-well plate to modified electrodes carrying mammalian cells .
Each isotype of C. perfringens carries a defined subset of virulence genes coding for toxin-producing sequences. Sergeev et al.  immobilized specific oligoprobes onto a multipathogen oligonucleotide microarray to detect six toxin-producing sequences in C. perfringens. Sequences encoding the different toxins hybridized strongly and specifically to the corresponding oligoprobes. The microarray-based test was applied on fluorescently-labeled amplicons obtain by initial PCR amplification step [81–83]. After hybridization, the microarray was scanned to measure the presence or absence of signal above background as an indicator of bacterial presence in the tested sample. This experimental approach was used for simultaneous analysis for several bacterial stains, their toxin genes or drug resistance determinants on a single-chip platform .
Finally, toxins produced by C. perfringens may be detected using specific antibodies integrated onto an immunosensor. Such a device was demonstrated for sensitive and label-free detection of epsilon toxin produced by C. perfringens . The sensor was obtained using an epsilon-toxin specific monoclonal antibody immobilized onto single walled carbon nanotubes. By controlling the morphology of the carbon nanotube assembly the sensor was adapted for detection of analytically relevant concentrations of toxin (nM range) and with sensitivities comparable to those of ELISA.
7 Detection of bluetongue and epizootic hemorrhagic disease viruses
Bluetongue a major non-contagious infectious disease of domestic and wild ruminants (mainly sheep, cattle, deer) is caused by the bluetongue virus which is an orbivirus of the Reoviridae family. Bluetongue virus is transmitted by the bite of a female of certain species of midges of the insect family Ceratopogonidae but can also infect embryos via the placenta or be transmitted via seminal fluid and colostrum . World Organisation for Animal Health (Office International des Epizooties, OIE) listed bluetongue virus because of its economic impact. The worldwide losses, estimated to 3 billion US$ a year , mainly affect ovine and bovine rearing industries. The estimated cost of bluetongue outbreaks in Scotland is £100 million per year (£30 million in direct losses and £70 million in indirect losses ), while the US losses in trade and associated testing of cattle for bluetongue virus status has been estimated at $130 million a year . It is worth noting that although the mortality and morbidity of bluetongue in cattle are rare; cattle can be infected by the virus without showing any clinical sign and in that way acting as reservoirs and contributing to the virus transmission. In consequence, economic losses come not only from direct damages caused by death, or reduced meat and milk production, but also from indirect effects due to the restriction in animal, semen or serum exporting. The clinical manifestations of bluetongue disease range from an unapparent or mild disease, to pyrexia, tachypnea, lethargy and even fatal disease .
In contrast to bluetongue virus which mainly affects sheep, strains of epizootic hemorrhagic disease virus, another distinct species within the genus Orbivirus, may exhibit high mortality and morbidity in both sheep and cows. As both diseases have similar clinical symptoms, both can be transmitted by the same insect and both have significant negative impacts on trade, the diagnosis of Orbiviruses is important. Current methods for bluetongue and epizootic hemorrhagic disease virus include numerous nucleic acid amplification assays, genotyping viral isolates, DNA microarray and next-generation sequencing . Promising advanced methods, such as fluorescent microsphere assays that can be adapted to single-tube multiplexing or multiple-well multiplexing, allow simultaneous detection of various viral RNA by in-solution hybridization . These tests are currently under validation in some laboratories. However, other alternative amplification methods, as loop-mediated isothermal amplification which can allow pen-site testing, or lateral flow test have still not be applied for Orbivirus detection. Nevertheless, a rapid lateral flow test for the detection of bluetongue virus-specific antibodies has been commercialized and recently validated .
8 Detection of Eimeria species
Intracellular protozoan parasite Eimeria causes coccidiosis which represents an important health problem in the poultry industry worldwide. The economic losses of the poultry industry due to coccidiosis are estimated to US$7 billion a year, mostly due to the late diagnosis because symptoms are not visible at the initial stages of infection. Eimeria undergoes rapidly its life cycle inside cells in intestinal tract of the infected bird. Upon finishing the internal phase in which the parasite is multiplied and excreted as oocysts in the faeces, it undergoes sporulation in the external phase and becomes infective. Infection in chicken leads to low feed conversion rate, reduced weight gain and increased mortality. At least seven different Eimeria species can affect poultry showing no cross-immunity between them . Usually under field conditions, coccidiosis is caused by infection with mixed but one dominant Eimeria specie. The damage occurring in the intestinal tract may facilitate secondary infection by some nonrelated bacteria, as documented for C. perfringens , or Salmonella Typhimurium [97, 98].
Traditional diagnosis employed in surveillance and control of coccidiosis is based on correlation of parasite size, morphology and site of infection with histological observation of lesions in infected birds. In addition to be complex and time consuming, these methods may also confuse coccidiosis with histomoniasis and salmonellosis due to their similar lesions. Biochemical and molecular diagnostic tools involve DNA based high-throughput analysis which permits to distinguish between distinct genetic parasites . New multiplex PCR assays, using primer pairs characteristic for all seven Eimeria causing infections in poultry, are performed on oocysts purified from the feces . Application of PCR-based assay for the presence of Eimeria oocysts on farms indicated limitations because of the assay low sensitivity (a minimum detection level was found to be 20 oocysts). This points out that classical PCR analysis cannot be applied for early infection diagnosis . Advances PRC technologies provide methods more adapted for early diagnosis as random amplification of polymorphic DNA PCR, DNA fingerprinting protocols and qPCR [102–104].
An alternative to classical PCR is a loop-mediated isothermal amplification (LAMP) method reported for the first time by Notomi et al. . LAMP is highly sensitive, specific and simple nucleic acid amplification method that functions at single constant temperature. It is also a time-saving method since an entire amplification is reached within less than 60 min. Overall, LAMP is quite adapted for elaboration of pathogens point-of-care diagnostic kits. LAMP is allowed by autocycling strand-displacement reaction using a set of oligonucleotides specific for different DNA sequences within the target genomic region and formation of a loop-structured amplicon. To improve amplification, additional loop primers may be added to the reaction. Barkway et al. [106, 107] have developed a panel of sensitive LAMP-based assays for diagnosis of seven Eimeria species that infect chickens. The test validation was performed on DNA extracted from mechanically disrupted Eimeria oocysts purified from faecal material. This study suggested that, although non-quantitative, LAMP-based diagnostics of coccidiosis parasite show many advantages: (i) the colorimetric read-out provided an instant result as no requirement for electrophoresis step; (ii) test sensitivity allowed parasite detection upon ongoing infection, and at early stages when any visible lesions can be resolved; (iii) LAMP was less sensitive to PCR inhibitors present in tested samples as some metal-ions or proteases compared to PCR and, finally, (iv) the completed test was cheaper than ~£1 per sample. Thus, simplicity, low cost and no requirement for sophisticated laboratory equipment make the LAMP-colorimetric paper test a good candidate for in field applications.
9 Detection of foot-and-mouth disease viruses
Foot-and-mouth disease virus causes an extremely infectious and contagious disease of cloven-hoofed animals as cattle, pigs, sheep and many wild species. Although showing a low mortality, foot-and-mouth disease has the global impact estimated between US$6.5 and 21 billion a year, due to high morbidity of the disease and the huge numbers of animals affected . It was estimated that one infected cow can infect more than 70 cattle, which makes foot-and-mouth virus the most infectious human and animal pathogen known. The clinical symptoms of food-and-mouth disease are formation of painful fluid-filled vesicles (blisters) or some erosion on the mouth tissues as tongue and lips, or other parts of the body where the skin is thin, like between the two toes of the feet. The pain causes many supplementary symptoms as depression, excessive salivation, lameness, anorexia and reluctance to move or stand.
The developed countries have eradicated the foot-and-mouth disease but widespread and long distance movements of animals sometimes re-incur virus. The outbreaks in disease free regions cause economic losses of about US$1.5 billion a year. Therefore a rapid implementation of the measures and a coordination between regions and countries are needed to control disease spread . However, even nowadays the virus identification cannot be done in all endemic regions. Many developing countries during the suspected outbreaks have to send infected samples to international laboratories because of the lack of resource and laboratory equipment. The transportation represents a major biosecurity risk, increases time of analysis and risk of sample degradation and has additional cost. In consequence, there is a need for a portable device for in field foot-and-mouth diagnosis to assure better disease control, surveillance and management in outbreaks.
The food-and-mouth virus belongs to the Aphthovirus genus of the Picornaviridae family. It consists of a single-stained positively-charged RNA genome surrounded by a capsid composed of four structural peptides VP1-4. There are seven immunologically and genetically distinct serotypes. There is no cross protection between different serotypes which implies that an outbreak diagnostic of both virus and of their serotype have to be done. Classical methods for the food-and-mouth diagnostics involve virus isolation, ELISA and RT-PCR tests . Recently an improved duplex one-step RT-PCR assays was validated . The main advantage of this test is in the co-amplification of foot-and-mouth virus RNA and host β-actin mRNA. Host mRNA served as an intern control to ensure accuracy and to avoid false negative results that occur when quality of RNA in samples is poor. Although the assay was validated as specific and sensitive, it is only adapted to laboratory diagnostics.
Fowler et al.  reported a lateral flow device for recovery of food-to-mouth viral RNA. In their assay the virus detection and subtype identification were performed in laboratory facilities while the lateral flow device was only proposed to improve field sample preservation during transport. To test the proof of concept, the authors applied epithelial suspension or cell culture infected with a food-to-mouth virus representing various serotypes on the antigen lateral flow strip. Specific antigen–antibody reaction resulted in development of the test line. This lateral flow device was then employed as a dry, thermos-stable and non-hazardous transport system of infected sample for subsequent nucleic acid amplification, sequencing or virus recovery in laboratory facilities.
Reid et al.  developed a rapid and simple chromatographic strip foot-and-mouth diagnostic test for field application. This device was shown to detect specifically foot-and-mouth disease virus antigen in laboratory and field condition. The chromatographic strip test contained a specific monoclonal antibody with broad reactivity for foot-and-mouth disease viruses. The device was validated with laboratory-based tests on a range of samples as nasal swabs, epithelial suspensions and probangs from clinical samples or from animals infected experimentally as well as in supernatant fluids resulting from their passage in cell culture.
Another example of a preliminary validated point-of-care test for foot-and-month disease diagnosis is a device that coupled reverse transcription loop-mediated isothermal amplification (RT-LAMP) with a lateral flow strips . The test provided diagnosis within less than 1 h with no requirement for instrumentation because the test line is visible by a naked eye. Moreover, the test can be performed in a standard water bath or heating block because RT-LAMP technique does not require a thermocycler as it amplifies specific nucleotide sequences at a constant temperature. The validation of the device has shown a successful detection of foot-and-month disease virus RNA from epithelial suspensions without the need for prior RNA extraction. All these advantages together with a possibility that a RT-LAMP method can be adapted for a high throughput system make this assay attractive for field use.
10 Detection of Campylobacter
Campylobacter is a Gram-negative spiral-shaped bacterium, mobile with flagella, which belongs to the Campylobacteriaceae family. It is microaerophilic and thermophilic microorganism that can grow well at the temperature range between 37 and 42 °C. Campylobacter can cause diseases in humans and animals including wild animals, pets and livestock species. C. jejuni, C. coli and C. fetus strains have been found worldwide. Particularly, C. jejuni and C. coli have been found in sheep, cattle, chickens, turkeys, dogs, cats and pigs while C. fetus has been found mostly in sheep, goats and cattle. C. jejuni, mostly found in poultry, may colonize the intestine of turkeys, chickens and waterfowl. Clinical signs of infection are enteritis with diarrhea, vomiting and fever, but also abortions and infertility, bovine genital campylobacteriosis, ileitis in hamsters, colitis in ferrets, enteritis in porcine. Lesions of infected young animals include edema of the mucosa of the ileum and cecum, mononuclear infiltration of the submucosa, villous atrophy with intraluminal accumulation of mucus. In addition, animals can be infected asymptomatically with any of Campylobacter strains. For instance, C. jejuni is not considered to be pathogenic in birds. Still, some cases of C. jejuni infection in chickens have been reported to cause enteritis and death.
Poultry is considered as the main source of human campylobacteriosis [114, 115], the disease caused by the infection with Campylobacter. The WHO recommended a control and surveillance of Campylobacter in poultry to reduce the risk for humans since this bacterium has a significant impact on public health. In fact, contaminated, undercooked poultry is responsible for 50–80% of cases of campylobacteriosis investigated. Nevertheless, contaminated beef and pork products are also responsible for some infections of people. Costs due to C. jejuni infections are estimated between US$1.5 billion to US$8.0 billion a year and about €2.4 billion a year, in the United States and Europe, respectively [116, 117].
The recommended procedure for detection of Campylobacter requires 4 days to provide response on its presence or absence, while up to 7 more days are required for Campylobacter strain identification. Microaerophilic conditions, a specific temperature, and selective enrichment media are required to grow Campylobacter in laboratory. However, classical plate based methods are not quite suitable for diagnostics Campylobacter. First, the strain differentiation between C. jejuni and C. coli is difficult using conventional cellular or biochemical methods, because of the similar characteristics of the two species . Second, transport can stress bacteria making them viable but-not culturable on selective agar plates, thus making plate count methods inefficient.
Molecular biology methods, that use nucleic acids as target for the tests allow detection of the viable but-not culturable microorganisms, reduce the time required to obtain results and improve specificity. Various kinds of specific and sensitive PCR-based tests have been developed that are usually run in laboratories. Thus, they require the transportation of the samples from the site of analysis to the equipped facilities. Regarding the sensitivity of Campylobacter to a variety of environmental conditions, the transport should be fast to reduce the loss of viability.
A portable device could help farmers to obtain a rapid detection of the pathogen, reducing the spread of the microorganism in the livestock and consequently of the food that could cause human illness. Barletta et al.  proposed a test based on melting-point curve analysis which identifies post-PCR products of C. jejuni. They standardized a classical PCR test with primers already reported against 16S rRNA of the Campylobacter spp. The test can be run using a multiple PCR for performing detection of Salmonella, Shigella and Campylobacter spp. Although selective, the method requires agarose gel migration of samples, making it slow and not adapted for screening high number of samples. In contrast, fluorescence-based RT-PCR methods are more robust and faster because no post-PCR protocols are required. Unfortunately, both protocols require the extraction of DNA and, thus, depend on the efficiency of the enzymatic reaction. As results may depend on the reagents used for enzymatic reaction, PCR-methods avoiding the amplification step would be more accurate.
Zhang et al.  described the utilization of magneto strictive particle coated with three layers of silica to bind anti-C. jejuni antibodies. Such functionalized magnetic particles were shown to detect C. jejuni in water released from chicken factories. Although bacterial cells are typically diluted in water, the easy manipulation of magnetic particles allowed bacterial cells concentration giving a sensitive detection with the detection limit of 102 CFU/mL. In another study, Wei et al.  proposed a biosensor based on surface plasmon resonance for identification of C. jejuni. Using specific antibodies against C. jejuni a sensitivity of 103 CFU/mL was reached.
Gnanaprakasa et al.  developed an optical biosensor for the detection of C. jejuni based on diffraction optic technology and surface plasmon resonance. The hippuricase gene (hipO) of C. jejuni was used as a target sequence. After being thiolated the complementary probe was immobilized on the gold surface of the sensorchip. On-chip hybridization was traduced to an optical signal which allowed C. jejuni quantification in infected samples with a detection limit of 5.0 nM.
Wald et al.  used a lateral flow method to detect C. jejuni or C. coli reaching a sensitivity of about 7.3 log of CFU/g. Though not highly sensitive this device is portable and can be adapted to multiplex detection. Possibility to have multi-pathogen in field diagnosis will significantly reduce time of detection allowing the activation of control measures in a short time. The improving of the hygiene of poultry will reduce the number of infections caused by Campylobacter in farms and thus the costs of human hospitalization.
11 Detection of Salmonella
Salmonella is a Gram-negative, rod-shaped, facultative anaerobic, non-spore-forming, motile enterobacteria, that can grow over the temperature range from 7 to 48 °C, and at optimum pH from 6.5 to 7.5. Salmonella is a cause of foodborne illness worldwide and is responsible of livestock infections that can be transmitted from animals to humans. Poultry are considered as one of the most important Salmonella reservoirs. Various serotypes of Salmonella overlap between farm animals and humans . Animals are mainly infected through feed, drinking water or environmental sources. Poultry farmers which suffers flock losses due to Salmonella infection, suffers also for eggs and other consumable poultry products losses. Various different strains of Salmonella have been identified although only two species have been recognized: S. enterica and S. bongori. Salmonella enterica can be subdivided in subspecies, Salmonella enterica subsp. enterica (I), Salmonella enterica subsp. salamae (II), Salmonella enterica subsp. arizonae (IIIa), Salmonella enterica subsp. diarizonae (IIIb), Salmonella enterica subsp. houtenae (IV), Salmonella enterica subsp. houtenae (VI). Salmonella bongori was originally designated as S. enterica subspecies V. The most important subspecies involved in poultry infections are Salmonella Typhimurium, Salmonella Enteritidis, Salmonella Virchow, Salmonella Hadar. Among all strains, S. Enteritidis is the serovar mostly implicated in salmonellosis worldwide. Affected birds are anorectic, weak, have diarrhea and may die shortly after hatching. Those who survive rest small in size and frequently carry localized infection of the ovary. In consequence survived hens may transmit the infection to the eggs and produce infected chicks.
The ISO 6579:2002 (AnnexD) method is currently the EU standard method for diagnosis of Salmonella in poultry samples. Other standard culture-based methods are available. Some of them have been validated by other regulatory agencies. Most of these conventional culture-based methods comprise the following steps: (i) liquid pre-enrichment on non-selective medium followed by that on (ii) liquid selective media containing inhibiting compounds towards non-Salmonella bacteria, (iii) culture on solid selective agar, and (iv) isolation of suspected colonies from the solid media, their incubation in a specific solid media followed by additional biochemical and serological tests for strain identification. All culture-based methods need complex laboratory equipment and at least 1 week to give results. Moreover, some serotypes cannot be detected on the selective media used for isolation, thus, false negatives can be provided by the test. For instance, Salmonella Virchow cannot be cultivated on agar.
In consequence, immunological and serological tests have been developed for Salmonella diagnosis. The identification step that uses biochemical assays detecting a specific biomarker has reduced the volume of the reagents allowing the simultaneous inoculum of many wells on the same strip leading to a reaction that can give the answer within 24 h. Many of such kits are commercially available (as API 20E, bioMerieux; Enterotube II, BD diagnostics, etc.). Nevertheless, it is difficult to incorporate immunoassays into an array to detect multiple targets due to the high cross-reactivity between antibodies. On the other side, ELISA, although high sensitive and fast, still can give false negative results, cross-reactivity problems, and need a pre-enrichment step .
The utilization of molecular biology and PCR-based methods, such RT-PCR and Digital-PCR, can reduce the time for Salmonella detection to several hours. These methods may be employed even when serotyping fails due to the change or loss of the surface antigens. Various kinds of rapid tests for Salmonella have been developed and approved, including the VIDAS Salmonella (SLM) method, the immuno-concentration Salmonella (ICS) method, the Tecra Unique Salmonella test, and the polymerase chain reaction PCR based BAX™ system, approved by the Association of Official Analytical Chemists (AOAC) International. The results are obtained within 24 h. Also Dynal BeadRetriever, Pathatrix, and BioVeris M1M Platform based on immunomagnetic separation (IMS) systems, and many others are currently used. Gene-Trak® (Neogen Corporation) bases on DNA probe hybridization  allows after the lysis of the target cells, the release of DNA for hybridization to the probe. Most molecular tests use an enrichment step to increase the sensitivity extending the time of the test and require supplementary equipment for the analysis.
In addition to food-borne pathogens detection and sanitary control of farms, the possible utilization of Salmonella for bioterrorism led to the development of portable biosensor-based detection methods. Biosensors may detect specific Salmonella proteins or DNA sequences in a rapid, specific, simple and sensitive way. Chai et al.  described a wireless magneto elastic mass-sensitive biosensor for the detection of S. Typhimurium, able to detect 1.6 × 102 CFU/mL. Wang et al.  reported the utilization of microcantilever decorated with peptides having high affinities to various strains of Salmonella able to detect only one cell of the pathogen in an infected sample. The cantilever bends lightly, due to the capture of a bacterium by the specific peptide, and a laser activates the alarm. The authors used phage-derived peptides to obtain an array for parallel detecting of eight serovars. Phage display uses genetic modification of viruses that infect bacteria, bacteriophages, to insert a sequence of the peptide that specifically recognizes one biomarker into a phage coat protein gene. In that way, the phage starts to express the peptide on its surface. The displaying phages can be attached on the biosensor surface and assure specific Salmonella detection.
A lateral flow system for Salmonella AD, produced by the DuPont, uses two-step enrichment (16–22 h plus 6–8 h) before the sample insertion on the strip. Strips contained colloidal gold conjugated anti-Salmonella antibodies coated onto the sample pad. The test provides results in 10 min. Strips have been validated for Salmonella detection on raw meat, chicken sera, eggs, processed meat, fruits, vegetables, etc. This method was also confirmed on meat samples spiked with 1–4 CFU/25 g before the enrichment.
Bulut  reported the development of a lateral flow immunochromatographic test platform for Salmonella using the invA gene as a target. This gene encodes the protein responsible for the invasivity of the pathogen. The platform requires no labeling of the target as it detects amplicons obtained by using specific primers followed by sandwich hybridization. Nevertheless, the method is not quantitative since it involves a PCR step. Fang et al.  developed a lateral flow biosensor based on aptamers to detect Salmonella. This method needed an amplification step to produce an amplified ssDNA before its deposition onto the sensor membrane. The reported sensitivity was 101 CFU/mL.
Microfluidic devices have also been used for the detection of Salmonella. Kim et al.  detected S. Typhimurium using a microfluidic device and specific anti-Salmonella antibodies able to efficiently capture Salmonella cells. S. Typhimurium was used to spike samples at various concentrations and establish the detection limit. A portable fluorometer using a LED unit detected the signal produced by the quantum dots conjugated to antibodies. The detection limit obtained for the sensor was 103 CFU/mL Salmonella in chicken meat extract.
12 Detection of bovine respiratory syncytial viruses
Bovine respiratory syncytial virus (BRSV) is spread worldwide and represents a major contributor of respiratory disease in cattle, especially in young beef and dairy cattle. The high prevalence of seropositive cattle infection is common in the cattle population making BRSV the most economically important and outstanding welfare issues for industrialized beef cattle producers and animal health organizations. The virus seems to spread by an aerosol route and can be easily transmitted by contact with respiratory secretions from infected cattle. Frequently, BRSV infection is associated with secondary bacterial infections that are treated by the massive use of antibiotics. In consequence, BRSV represents an additional public health-concern through the risk of antibiotic-resistance developing.
BRSV is a single-strained, negative-sense RNA enveloped virus classified as a Pneumovirus of Paramyxoviridae family. The genome of BRSV encodes for 11 proteins: nucleocapsid (N), phosphoprotein, large polymerase, transcriptional antitermination factor M2-1 and RNA regulatory protein M2-2 that all associate the genomic RNA, then attachment (G), fusion (F) and small hydrophobic protein that are transmembrane surface glycoproteins, while matrix or membrane protein (M) is a nonglycosylated and associates with the inner face of the envelope. In addition, two nonstructural proteins, named NS1 and NS2, accumulate within the infected cells.
After the incubation period estimated to take between 2 and 5 days, BRSV infection is usually limited to the upper respiratory tract giving clinical signs that may range from minimal to severe with dyspnea and death. Affected calves can have tachypnea, ocular serous secretions, dry muzzle, reduced activity, anorexia, and high fevers. The clinical signs may last until 7–10 days post-infection. Interestingly, the antibody response is frequently developed few days before clinical signs appear. Thus, testing serum samples pooled from a number of animals in a respiratory outbreak may help diagnosis. Still, although clinical signs may suggest BRSV infection, diagnosis of BRSV requires a laboratory confirmation.
PCR is commonly used as a molecular method for BRSV diagnosis. However, the virus isolation is difficult, except in some cases upon the acute phases of infection. Detection of negative-sense RNA genome or mRNA of the virus is even more delicate than detection of viral proteins. First, RNAs are sensitive to RNAse digestion and degradation and have shorter life time than proteins. Second, RSV mRNA is expressed cyclically, while protein expression is stable and increases in a regular manner over time. In consequence, ELISA test that quantify the RSV fusion protein subunits F0 and F1, expressed by all RSV strains, provides an efficient diagnostic tool. Similarly, the detection of specific antibodies in serum samples has proved useful to establish a diagnosis of BRSV antigen in the acute sample. Interestingly, the antibody titer in animals with developed clinical signs of the disease is usually higher than in the sample taken several weeks later.
Cai et al.  proposed a label-free electrochemical biosensor based on molecular beacon for detection of target mRNA from RSV. Molecular beacons are oligonucleotide hybridization probes in a form of a hairpin shaped molecules. Within the hairpin structure a streptavidin binding aptamer sequence was blocked for this assay. Upon hybridization of the beacon with specific nucleic acid sequence, the hairpin opens to allow binding to streptavidin-HRP protein complex. In that way, the hybridization event can be quantified by the enzymatic reaction using a HRP substrate, tetramethylbenzidine (TMB). The sensor was shown to detect 13 amol of RSV mRNA in a 4 µL total volume of a complex biological media. The high specificity and selectivity of the device was demonstrated as sensor detected specific RSV mRNA but not one base mismatched sequence.
In another study, the HRP-TMB reaction was used for electrochemical detection of RSV specific antigens . For this, the polystyrene array slide that allows antibody immobilization was attached with disposable screen-printed electrodes. The RSV fusion protein was captured in a sandwich by a specific monoclonal antibody attached to the polystyrene surface and a second antibody-coupled with HRP. The enzymatic reaction HRP-TMB was traduced into an amperometric signal by the electrode. The whole test needed only 25 min to produce result showing similar selectivity and sensibility to commercial RT-PCR or immunofluorescent tests when probed on clinical samples.
Perez et al.  described the detection of RSV by nanoparticle amplification of immuno-PCR assay. Their test detects RSV by viral particles capturing into a sandwich formed between two different antibodies specific against the F protein. The first antibody was attached to magnetic beads to allow extraction, while the second was co-immobilized onto gold nanoparticles with specific DNA sequences. After extraction the hybridized DNA can be released by heating and detected via RT-PCR. It was shown that this nanoparticle amplified immune-PCR assay not only provided better virus sampling but also reduce the effect of PCR-induced variation in sample replicates by increasing the tag DNA concentration and by lowering the background signaling. Compared to ELISA and RT-PCR detection, the nanoparticle amplified immune-PCR showed a several 1000-fold improvement in the limit of detection as it detected 4.1 PFU/mL.
The authors declare that they have no competing interests.
JV designed the structure of the review and took the lead in drafting of the manuscript. NJ, MM and CMC drafted several sections of the manuscript and helped with the revision of the draft. All authors read and approved the final manuscript.
This work was funded through the grant of the Agence Nationale de la Recherche SENSAIV (ANR-15-MRSE-0028-01) to JV, BioAsia Grant 35976PH to JV, and MOST Grant, Southern Taiwan Medical Device Industry Cluster (CY-08-04-22-105) Taiwan to CMC.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.
- Kuiken T, Leighton FA, Fouchier RA, LeDuc JW, Peiris JS, Schudel A, Stohr K, Osterhaus AD (2005) Public health. Pathogen surveillance in animals. Science 309:1680–1681PubMedView ArticleGoogle Scholar
- Yoo SM, Lee SY (2016) Optical biosensors for the detection of pathogenic microorganisms. Trends Biotechnol 34:7–25PubMedView ArticleGoogle Scholar
- Vashist SK, Lam E, Hrapovic S, Male KB, Luong JH (2014) Immobilization of antibodies and enzymes on 3-aminopropyltriethoxysilane-functionalized bioanalytical platforms for biosensors and diagnostics. Chem Rev 114:11083–11130PubMedView ArticleGoogle Scholar
- Fournier-Wirth C, Jaffrezic-Renault N, Coste J (2010) Detection of blood-transmissible agents: can screening be miniaturized? Transfusion 50:2032–2045PubMedView ArticleGoogle Scholar
- Bahadır EB, Sezgintürk MK (2015) Applications of commercial biosensors in clinical, food, environmental, and biothreat/biowarfare analyses. Anal Biochem 478:107–120PubMedView ArticleGoogle Scholar
- Zhu C, Yang G, Li H, Du D, Lin Y (2014) Electrochemical sensors and biosensors based on nanomaterials and nanostructures. Anal Chem 87:230–249PubMedPubMed CentralView ArticleGoogle Scholar
- Karch H, Denamur E, Dobrindt U, Finlay BB, Hengge R, Johannes L, Ron EZ, Tonjum T, Sansonetti PJ, Vicente M (2012) The enemy within us: lessons from the 2011 European Escherichia coli O104:H4 outbreak. EMBO Mol Med 4:841–848PubMedPubMed CentralView ArticleGoogle Scholar
- Belanger L, Garenaux A, Harel J, Boulianne M, Nadeau E, Dozois CM (2011) Escherichia coli from animal reservoirs as a potential source of human extraintestinal pathogenic E. coli. FEMS Immunol Med Microbiol 62:1–10PubMedView ArticleGoogle Scholar
- Hobson NS, Tothill I, Turner AP (1996) Microbial detection. Biosens Bioelectron 11:455–477PubMedView ArticleGoogle Scholar
- Fournier-Wirth C, Deschaseaux M, Defer C, Godreuil S, Carrière C, Bertrand X, Tunez V, Schneider T, Coste J, Morel P (2006) Evaluation of the enhanced bacterial detection system for screening of contaminated platelets. Transfusion 46:220–224PubMedView ArticleGoogle Scholar
- Dharmasiri U, Witek MA, Adams AA, Osiri JK, Hupert ML, Bianchi TS, Roelke DL, Soper SA (2010) Enrichment and detection of Escherichia coli O157: H7 from water samples using an antibody modified microfluidic chip. Anal Chem 82:2844–2849PubMedPubMed CentralView ArticleGoogle Scholar
- Jiang X, Wang R, Wang Y, Su X, Ying Y, Wang J, Li Y (2011) Evaluation of different micro/nanobeads used as amplifiers in QCM immunosensor for more sensitive detection of E. coli O157: H7. Biosens Bioelectron 29:23–28PubMedView ArticleGoogle Scholar
- Yazgan I, Noah NM, Toure O, Zhang S, Sadik OA (2014) Biosensor for selective detection of E. coli in spinach using the strong affinity of derivatized mannose with fimbrial lectin. Biosens Bioelectron 61:266–273PubMedView ArticleGoogle Scholar
- Zhang Y, Tan C, Fei R, Liu X, Zhou Y, Chen J, Chen H, Zhou R, Hu Y (2014) Sensitive chemiluminescence immunoassay for E. coli O157: H7 detection with signal dual-amplification using glucose oxidase and laccase. Anal Chem 86:1115–1122PubMedView ArticleGoogle Scholar
- Guo Y, Wang Y, Liu S, Yu J, Wang H, Wang Y, Huang J (2016) Label-free and highly sensitive electrochemical detection of E. coli based on rolling circle amplifications coupled peroxidase-mimicking DNAzyme amplification. Biosens Bioelectron 75:315–319PubMedView ArticleGoogle Scholar
- Kim M, Jung T, Kim Y, Lee C, Woo K, Seol JH, Yang S (2015) A microfluidic device for label-free detection of Escherichia coli in drinking water using positive dielectrophoretic focusing, capturing, and impedance measurement. Biosens Bioelectron 74:1011–1015PubMedView ArticleGoogle Scholar
- Idil N, Hedström M, Denizli A, Mattiasson B (2017) Whole cell based microcontact imprinted capacitive biosensor for the detection of Escherichia coli. Biosens Bioelectron 87:807–815PubMedView ArticleGoogle Scholar
- Jaffrezic-Renault N, Martelet C, Chevolot Y, Cloarec JP (2007) Biosensors and bio-bar code assays based on biofunctionalized magnetic microbeads. Sensors 7:589–614PubMed CentralView ArticleGoogle Scholar
- El Ichi S, Leon F, Vossier L, Marchandin H, Errachid A, Coste J, Jaffrezic-Renault N, Fournier-Wirth C (2014) Microconductometric immunosensor for label-free and sensitive detection of Gram-negative bacteria. Biosens Bioelectron 54:378–384PubMedView ArticleGoogle Scholar
- Eltzov E, Marks RS (2016) Miniaturized flow stacked immunoassay detects E. coli in a single step. Anal Chem 88:6441–6449PubMedView ArticleGoogle Scholar
- Eltzov E, Marks RS (2017) Colorimetric stack pad immunoassay for bacterial identification. Biosens Bioelectron 87:572–578PubMedView ArticleGoogle Scholar
- Olsen B, Munster VJ, Wallensten A, Waldenstrom J, Osterhaus AD, Fouchier RA (2006) Global patterns of influenza a virus in wild birds. Science 312:384–388PubMedView ArticleGoogle Scholar
- Morse SS, Mazet JA, Woolhouse M, Parrish CR, Carroll D, Karesh WB, Zambrana-Torrelio C, Lipkin WI, Daszak P (2012) Prediction and prevention of the next pandemic zoonosis. Lancet 380:1956–1965PubMedPubMed CentralView ArticleGoogle Scholar
- First European outbreak of avian influenza A(H5N8) confirmed in poultry. http://www.euro.who.int/en/health-topics/communicable-diseases/influenza/news/news/2014/11/first-european-outbreak-of-avian-influenza-ah5n8-confirmed-in-poultry. Accessed 26 Jan 2017
- Swayne DE (2009) Avian influenza. Wiley, New YorkGoogle Scholar
- Khanna M, Kumar P, Choudhary K, Kumar B, Vijayan V (2008) Emerging influenza virus: a global threat. J Biosci 33:475–482PubMedView ArticleGoogle Scholar
- Vasin AV, Temkina OA, Egorov VV, Klotchenko SA, Plotnikova MA, Kiselev OI (2014) Molecular mechanisms enhancing the proteome of influenza A viruses: an overview of recently discovered proteins. Virus Res 185:53–63PubMedView ArticleGoogle Scholar
- Jagger BW, Wise HM, Kash JC, Walters KA, Wills NM, Xiao YL, Dunfee RL, Schwartzman LM, Ozinsky A, Bell GL, Dalton RM, Lo A, Efstathiou S, Atkins JF, Firth AE, Taubenberger JK, Digard P (2012) An overlapping protein-coding region in influenza A virus segment 3 modulates the host response. Science 337:199–204PubMedPubMed CentralView ArticleGoogle Scholar
- Muramoto Y, Noda T, Kawakami E, Akkina R, Kawaoka Y (2013) Identification of novel influenza A virus proteins translated from PA mRNA. J Virol 87:2455–2462PubMedPubMed CentralView ArticleGoogle Scholar
- Selman M, Dankar SK, Forbes NE, Jia JJ, Brown EG (2012) Adaptive mutation in influenza A virus non-structural gene is linked to host switching and induces a novel protein by alternative splicing. Emerg Microbes Infect 1:e42PubMedPubMed CentralView ArticleGoogle Scholar
- Wise HM, Foeglein A, Sun J, Dalton RM, Patel S, Howard W, Anderson EC, Barclay WS, Digard P (2009) A complicated message: identification of a novel PB1-related protein translated from influenza A virus segment 2 mRNA. J Virol 83:8021–8031PubMedPubMed CentralView ArticleGoogle Scholar
- Chang CM, Lebarbenchon C, Gauthier-Clerc M, Le Bohec C, Beaune D, Le Maho Y, Van Der Werf S (2009) Molecular surveillance for avian influenza A virus in king penguins (Aptenodytes patagonicus). Polar Biol 32:663–665View ArticleGoogle Scholar
- Chiou CC, Chen TL, Tsao KC, Shih SR, Huang CG, Huang YL, Chang CM (2010) Detection of pandemic (H1N1) 2009 influenza virus by allele discrimination. Clin Chim Acta 411:1080–1083PubMedView ArticleGoogle Scholar
- Tseng CH, Tsai HJ, Chang CM (2014) A complete molecular diagnostic procedure for applications in surveillance and subtyping of avian influenza virus. Biomed Res Int 2014:653056PubMedPubMed CentralGoogle Scholar
- Baas C, Barr I, Fouchier R, Kelso A, Hurt A (2013) Clinical severity of human infections with avian influenza 2013. Eurosurveillance 18:20487PubMedGoogle Scholar
- Cho CH, Woo MK, Kim JY, Cheong S, Lee CK, An SA, Lim CS, Kim WJ (2013) Evaluation of five rapid diagnostic kits for influenza A/B virus. J Virol Methods 187:51–56PubMedView ArticleGoogle Scholar
- Hideshima S, Hinou H, Ebihara D, Sato R, Kuroiwa S, Nakanishi T, Nishimura S, Osaka T (2013) Attomolar detection of influenza A virus hemagglutinin human H1 and avian H5 using glycan-blotted field effect transistor biosensor. Anal Chem 85:5641–5644PubMedView ArticleGoogle Scholar
- Hidari KI, Shimada S, Suzuki Y, Suzuki T (2007) Binding kinetics of influenza viruses to sialic acid-containing carbohydrates. Glycoconj J 24:583–590PubMedView ArticleGoogle Scholar
- Suenaga E, Mizuno H, Penmetcha KK (2012) Monitoring influenza hemagglutinin and glycan interactions using surface plasmon resonance. Biosens Bioelectron 32:195–201PubMedView ArticleGoogle Scholar
- Gopinath SC, Awazu K, Fujimaki M, Shimizu K (2013) Neu5Acα2, 6Gal and Neu5Acα2, 3Gal receptor specificities on influenza viruses determined by a waveguide-mode sensor. Acta Biomater 9:5080–5087PubMedView ArticleGoogle Scholar
- Takahashi T, Kawagishi S, Masuda M, Suzuki T (2013) Binding kinetics of sulfatide with influenza A virus hemagglutinin. Glycoconj J 30:709–716PubMedView ArticleGoogle Scholar
- Lee C, Gaston MA, Weiss AA, Zhang P (2013) Colorimetric viral detection based on sialic acid stabilized gold nanoparticles. Biosens Bioelectron 42:236–241PubMedView ArticleGoogle Scholar
- Dinh H, Zhang X, Sweeney J, Yang Y, He Y, Dhawane A, Iyer SS (2014) Glycan based detection and drug susceptibility of influenza virus. Anal Chem 86:8238–8244PubMedPubMed CentralView ArticleGoogle Scholar
- Sajid M, Kawde AN, Daud M (2015) Designs, formats and applications of lateral flow assay: a literature review. J Saudi Chem Soc 19:689–705View ArticleGoogle Scholar
- Huang X, Aguilar ZP, Xu H, Lai W, Xiong Y (2016) Membrane-based lateral flow immunochromatographic strip with nanoparticles as reporters for detection: a review. Biosens Bioelectron 75:166–180PubMedView ArticleGoogle Scholar
- Li X, Lu D, Sheng Z, Chen K, Guo X, Jin M, Han H (2012) A fast and sensitive immunoassay of avian influenza virus based on label-free quantum dot probe and lateral flow test strip. Talanta 100:1–6PubMedView ArticleGoogle Scholar
- Wu F, Yuan H, Zhou C, Mao M, Liu Q, Shen H, Cen Y, Qin Z, Ma L, Li LS (2016) Multiplexed detection of influenza A virus subtype H5 and H9 via quantum dot-based immunoassay. Biosens Bioelectron 77:464–470PubMedView ArticleGoogle Scholar
- Abd El Wahed A, Weidmann M, Hufert FT (2015) Diagnostics-in-a-suitcase: development of a portable and rapid assay for the detection of the emerging avian influenza A (H7N9) virus. J Clin Virol 69:16–21PubMedView ArticleGoogle Scholar
- Nidzworski D, Pranszke P, Grudniewska M, Krol E, Gromadzka B (2014) Universal biosensor for detection of influenza virus. Biosens Bioelectron 59:239–242PubMedView ArticleGoogle Scholar
- Hewa TMP, Tannock GA, Mainwaring DE, Harrison S, Fecondo JV (2009) The detection of influenza A and B viruses in clinical specimens using a quartz crystal microbalance. J Virol Methods 162:14–21View ArticleGoogle Scholar
- Miodek A, Sauriat-Dorizon H, Chevalier C, Delmas B, Vidic J, Korri-Youssoufi H (2014) Direct electrochemical detection of PB1-F2 protein of influenza A virus in infected cells. Biosens Bioelectron 59:6–13PubMedView ArticleGoogle Scholar
- Miodek A, Vidic J, Sauriat-Dorizon H, Richard CA, Le Goffic R, Korri-Youssoufi H, Chevalier C (2014) Electrochemical detection of the oligomerization of PB1-F2 influenza A virus protein in infected cells. Anal Chem 86:9098–9105PubMedView ArticleGoogle Scholar
- Su LC, Chang CM, Tseng YL, Chang YF, Li YC, Chang YS, Chou C (2012) Rapid and highly sensitive method for influenza A (H1N1) virus detection. Anal Chem 84:3914–3920PubMedView ArticleGoogle Scholar
- Fu Y, Callaway Z, Lum J, Wang R, Lin J, Li Y (2014) Exploiting enzyme catalysis in ultra-low ion strength media for impedance biosensing of avian influenza virus using a bare interdigitated electrode. Anal Chem 86:1965–1971PubMedView ArticleGoogle Scholar
- Wang R, Zhao J, Jiang T, Kwon YM, Lu H, Jiao P, Liao M, Li Y (2013) Selection and characterization of DNA aptamers for use in detection of avian influenza virus H5N1. J Virol Methods 189:362–369PubMedView ArticleGoogle Scholar
- Cui ZQ, Ren Q, Wei HP, Chen Z, Deng JY, Zhang ZP, Zhang XE (2011) Quantum dot-aptamer nanoprobes for recognizing and labeling influenza A virus particles. Nanoscale 3:2454–2457PubMedView ArticleGoogle Scholar
- Cheng C, Dong J, Yao L, Chen A, Jia R, Huan L, Guo J, Shu Y, Zhang Z (2008) Potent inhibition of human influenza H5N1 virus by oligonucleotides derived by SELEX. Biochem Biophys Res Commun 366:670–674PubMedView ArticleGoogle Scholar
- Jeon SH, Kayhan B, Ben-Yedidia T, Arnon R (2004) A DNA aptamer prevents influenza infection by blocking the receptor binding region of the viral hemagglutinin. J Biol Chem 279:48410–48419PubMedView ArticleGoogle Scholar
- Li J, Chen S, Evans DH (2001) Typing and subtyping influenza virus using DNA microarrays and multiplex reverse transcriptase PCR. J Clin Microbiol 39:696–704PubMedPubMed CentralView ArticleGoogle Scholar
- Kessler N, Ferraris O, Palmer K, Marsh W, Steel A (2004) Use of the DNA flow-thru chip, a three-dimensional biochip, for typing and subtyping of influenza viruses. J Clin Microbiol 42:2173–2185PubMedPubMed CentralView ArticleGoogle Scholar
- Wang D, Coscoy L, Zylberberg M, Avila PC, Boushey HA, Ganem D, DeRisi JL (2002) Microarray-based detection and genotyping of viral pathogens. Proc Natl Acad Sci U S A 99:15687–15692PubMedPubMed CentralView ArticleGoogle Scholar
- Townsend MB, Dawson ED, Mehlmann M, Smagala JA, Dankbar DM, Moore CL, Smith CB, Cox NJ, Kuchta RD, Rowlen KL (2006) Experimental evaluation of the FluChip diagnostic microarray for influenza virus surveillance. J Clin Microbiol 44:2863–2871PubMedPubMed CentralView ArticleGoogle Scholar
- Liu RH, Lodes MJ, Nguyen T, Siuda T, Slota M, Fuji HS, McShea A (2006) Validation of a fully integrated microfluidic array device for influenza A subtype identification and sequencing. Anal Chem 78:4184–4193PubMedView ArticleGoogle Scholar
- Caswell J, Archambault M (2007) Mycoplasma bovis pneumonia in cattle. Anim Health Res Rev 8:161–186PubMedView ArticleGoogle Scholar
- Moon JS, Lee AR, Kang HM, Lee ES, Joo YS, Park YH, Kim MN, Koo HC (2007) Antibiogram and coagulase diversity in staphylococcal enterotoxin-producing Staphylococcus aureus from bovine mastitis. J Dairy Sci 90:1716–1724PubMedView ArticleGoogle Scholar
- Sargeant JM, Leslie KE, Shirley JE, Pulkrabek BJ, Lim GH (2001) Sensitivity and specificity of somatic cell count and California Mastitis Test for identifying intramammary infection in early lactation. J Dairy Sci 84:2018–2024PubMedView ArticleGoogle Scholar
- Pemberton RM, Hart JP, Mottram TT (2001) An assay for the enzyme N-acetyl-beta-d-glucosaminidase (NAGase) based on electrochemical detection using screen-printed carbon electrodes (SPCEs). Analyst 126:1866–1871PubMedView ArticleGoogle Scholar
- Akerstedt M, Bjorck L, Persson Waller K, Sternesjo A (2006) Biosensor assay for determination of haptoglobin in bovine milk. J Dairy Res 73:299–305PubMedView ArticleGoogle Scholar
- Fu P, Sun Z, Yu Z, Zhang Y, Shen J, Zhang H, Xu W, Jiang F, Chen H, Wu W (2014) Enzyme linked aptamer assay: based on a competition format for sensitive detection of antibodies to Mycoplasma bovis in serum. Anal Chem 86:1701–1709PubMedView ArticleGoogle Scholar
- Koskinen MT, Holopainen J, Pyorala S, Bredbacka P, Pitkala A, Barkema HW, Bexiga R, Roberson J, Solverod L, Piccinini R, Kelton D, Lehmusto H, Niskala S, Salmikivi L (2009) Analytical specificity and sensitivity of a real-time polymerase chain reaction assay for identification of bovine mastitis pathogens. J Dairy Sci 92:952–959PubMedView ArticleGoogle Scholar
- Pisoni G, Moroni P, Genini S, Stella A, Boettcher PJ, Cremonesi P, Scaccabarozzi L, Giuffra E, Castiglioni B (2010) Differentially expressed genes associated with Staphylococcus aureus mastitis in dairy goats. Vet Immunol Immunopathol 135:208–217PubMedView ArticleGoogle Scholar
- Lee KH, Lee JW, Wang SW, Liu LY, Lee MF, Chuang ST, Shy YM, Chang CL, Wu MC, Chi CH (2008) Development of a novel biochip for rapid multiplex detection of seven mastitis-causing pathogens in bovine milk samples. J Vet Diagn Investig 20:463–471View ArticleGoogle Scholar
- Mujawar LH, Moers A, Norde W, van Amerongen A (2013) Rapid mastitis detection assay on porous nitrocellulose membrane slides. Anal Bioanal Chem 405:7469–7476PubMedView ArticleGoogle Scholar
- Braiek M, Rokbani KB, Chrouda A, Mrabet B, Bakhrouf A, Maaref A, Jaffrezic-Renault N (2012) An electrochemical immunosensor for detection of Staphylococcus aureus bacteria based on immobilization of antibodies on self-assembled monolayers-functionalized gold electrode. Biosensors 2:417–426PubMedPubMed CentralView ArticleGoogle Scholar
- Gibert M, Jolivet-Renaud C, Popoff MR (1997) Beta2 toxin, a novel toxin produced by Clostridium perfringens. Gene 203:65–73PubMedView ArticleGoogle Scholar
- Petit L, Gibert M, Popoff MR (1999) Clostridium perfringens: toxinotype and genotype. Trends Microbiol 7:104–110PubMedView ArticleGoogle Scholar
- Uzal F, Vidal J, McClane B, Gurjar A (2010) Clostridium perfringens toxins involved in mammalian veterinary diseases. Open Toxinol J 2:24View ArticleGoogle Scholar
- Keyburn AL, Boyce JD, Vaz P, Bannam TL, Ford ME, Parker D, Di Rubbo A, Rood JI, Moore RJ (2008) NetB, a new toxin that is associated with avian necrotic enteritis caused by Clostridium perfringens. PLoS Pathog 4:e26PubMedPubMed CentralView ArticleGoogle Scholar
- Timbermont L, Haesebrouck F, Ducatelle R, Van Immerseel F (2011) Necrotic enteritis in broilers: an updated review on the pathogenesis. Avian Pathol 40:341–347PubMedView ArticleGoogle Scholar
- Banerjee P, Bhunia AK (2010) Cell-based biosensor for rapid screening of pathogens and toxins. Biosens Bioelectron 26:99–106PubMedView ArticleGoogle Scholar
- Sergeev N, Distler M, Courtney S, Al-Khaldi SF, Volokhov D, Chizhikov V, Rasooly A (2004) Multipathogen oligonucleotide microarray for environmental and biodefense applications. Biosens Bioelectron 20:684–698PubMedView ArticleGoogle Scholar
- Al-Khaldi SF, Villanueva D, Chizhikov V (2004) Identification and characterization of Clostridium perfringens using single target DNA microarray chip. Int J Food Microbiol 91:289–296PubMedView ArticleGoogle Scholar
- Loy A, Bodrossy L (2006) Highly parallel microbial diagnostics using oligonucleotide microarrays. Clin Chim Acta 363:106–119PubMedView ArticleGoogle Scholar
- Volokhov DV, Kong H, Herold K, Chizhikov VE, Rasooly A (2011) Oligonucleotide microarrays for identification of microbial pathogens and detection of their virulence-associated or drug-resistance determinants. Methods Mol Biol 671:55–94PubMedView ArticleGoogle Scholar
- Palaniappan A, Goh W, Fam D, Rajaseger G, Chan C, Hanson B, Moochhala S, Mhaisalkar S, Liedberg B (2013) Label-free electronic detection of bio-toxins using aligned carbon nanotubes. Biosens Bioelectron 43:143–147PubMedView ArticleGoogle Scholar
- Sperlova A, Zendulkova D (2011) Bluetongue: a review. Vet Med Czech 56:430–452Google Scholar
- Tabachnick WJ (1996) Culicoides variipennis and bluetongue-virus epidemiology in the United States. Annu Rev Entomol 41:23–43PubMedView ArticleGoogle Scholar
- Assessing the economic impact of different bluetongue virus (BTV) incursion Scenarios in Scotland. http://www.gov.scot/Resource/Doc/241130/0066946.pdf. Accessed 26 Jan 2017
- Bluetongue. http://edis.ifas.ufl.edu/in768. Accessed 26 Jan 2017
- Schwartz-Cornil I, Mertens PP, Contreras V, Hemati B, Pascale F, Bréard E, Mellor PS, MacLachlan NJ, Zientara S (2008) Bluetongue virus: virology, pathogenesis and immunity. Vet Res 39:46PubMedView ArticleGoogle Scholar
- Wilson WC, Daniels P, Ostlund EN, Johnson DE, Oberst RD, Hairgrove TB, Mediger J, McIntosh MT (2015) Diagnostic tools for bluetongue and epizootic hemorrhagic disease viruses applicable to North American veterinary diagnosticians. Vector Borne Zoonotic Dis 15:364–373PubMedView ArticleGoogle Scholar
- Weis A, Liang F, Gao J, Barnard RT, Corrie S (2015) RNA and DNA diagnostics on microspheres: current and emerging methods. In: RNA DNA diagnostics, Springer, Heidelberg, pp 205–224Google Scholar
- Hanon JB, Vandenberge V, Deruelle M, De Leeuw I, De Clercq K, Van Borm S, Koenen F, Liu L, Hoffmann B, Batten CA (2016) Inter-laboratory evaluation of the performance parameters of a lateral flow test device for the detection of bluetongue virus-specific antibodies. J Virol Methods 228:140–150PubMedView ArticleGoogle Scholar
- Danielli A, Porat N, Arie A, Ehrlich M (2009) Rapid homogenous detection of the Ibaraki virus NS3 cDNA at picomolar concentrations by magnetic modulation. Biosens Bioelectron 25:858–863PubMedView ArticleGoogle Scholar
- Frölich S, Farhat J, Wallach M (2013) Designing strategies for the control of coccidiosis in chickens on poultry farms using modern diagnostic tools. Rep Parasitol 3:1–10Google Scholar
- Helmboldt C, Bryant E (1971) The pathology of necrotic enteritis in domestic fowl. Avian Dis 15:775–780PubMedView ArticleGoogle Scholar
- Arakawa A, Baba E, Fukata T (1981) Eimeria tenella infection enhances Salmonella typhimurium infection in chickens. Poult Sci 60:2203–2209PubMedView ArticleGoogle Scholar
- Baba E, Fukata T, Arakawa A (1982) Establishment and persistence of Salmonella typhimurium infection stimulated by Eimeria tenella in chickens. Res Vet Sci 33:95–98PubMedGoogle Scholar
- Gasser RB, Woods WG, Wood JM, Ashdown L, Richards G, Whithear KG (2001) Automated, fluorescence-based approach for the specific diagnosis of chicken coccidiosis. Electrophoresis 22:3546–3550PubMedView ArticleGoogle Scholar
- Fernandez S, Pagotto A, Furtado M, Katsuyama A, Madeira A, Gruber A (2003) A multiplex PCR assay for the simultaneous detection and discrimination of the seven Eimeria species that infect domestic fowl. Parasitology 127:317–325PubMedView ArticleGoogle Scholar
- Carvalho F, Wenceslau A, Teixeira M, Albuquerque G (2011) Molecular diagnosis of Eimeria species affecting naturally infected Gallus gallus. Genet Mol Res 10:996–1005PubMedView ArticleGoogle Scholar
- Morris G, Gasser R (2006) Biotechnological advances in the diagnosis of avian coccidiosis and the analysis of genetic variation in Eimeria. Biotechnol Adv 24:590–603PubMedView ArticleGoogle Scholar
- Beck HP, Blake D, Dardé ML, Felger I, Pedraza-Díaz S, Regidor-Cerrillo J, Gómez-Bautista M, Ortega-Mora LM, Putignani L, Shiels B (2009) Molecular approaches to diversity of populations of apicomplexan parasites. Int J Parasitol 39:175–189PubMedView ArticleGoogle Scholar
- Vrba V, Blake DP, Poplstein M (2010) Quantitative real-time PCR assays for detection and quantification of all seven Eimeria species that infect the chicken. Vet Parasitol 174:183–190PubMedView ArticleGoogle Scholar
- Notomi T, Okayama H, Masubuchi H, Yonekawa T, Watanabe K, Amino N, Hase T (2000) Loop-mediated isothermal amplification of DNA. Nucleic Acids Res 28:e63–e63PubMedPubMed CentralView ArticleGoogle Scholar
- Barkway CP, Pocock RL, Vrba V, Blake DP (2011) Loop-mediated isothermal amplification (LAMP) assays for the species-specific detection of Eimeria that infect chickens. BMC Vet Res 7:67PubMedPubMed CentralView ArticleGoogle Scholar
- Barkway CP, Pocock RL, Vrba V, Blake DP (2015) Loop-mediated isothermal amplification (LAMP) assays for the species-specific detection of Eimeria that infect chickens. J Vis Exp 96:52552Google Scholar
- Knight-Jones T, Rushton J (2013) The economic impacts of foot and mouth disease—what are they, how big are they and where do they occur? Prev Vet Med 112:161–173PubMedPubMed CentralView ArticleGoogle Scholar
- Jamal SM, Belsham GJ (2013) Foot-and-mouth disease: past, present and future. Vet Res 44:116PubMedPubMed CentralView ArticleGoogle Scholar
- Gorna K, Relmy A, Romey A, Zientara S, Blaise-Boisseau S, Bakkali-Kassimi L (2016) Establishment and validation of two duplex one-step real-time RT-PCR assays for diagnosis of foot-and-mouth disease. J Virol Methods 235:168–175PubMedView ArticleGoogle Scholar
- Fowler V, Bashiruddin JB, Belsham GJ, Stenfeldt C, Bøtner A, Knowles NJ, Bankowski B, Parida S, Barnett P (2014) Characteristics of a foot-and-mouth disease virus with a partial VP1 GH loop deletion in experimentally infected cattle. Vet Microbiol 169:58–66PubMedView ArticleGoogle Scholar
- Reid SM, Ferris NP, Brüning A, Hutchings GH, Kowalska Z, Åkerblom L (2001) Development of a rapid chromatographic strip test for the pen-side detection of foot-and-mouth disease virus antigen. J Virol Methods 96:189–202PubMedView ArticleGoogle Scholar
- Waters RA, Fowler VL, Armson B, Nelson N, Gloster J, Paton DJ, King DP (2014) Preliminary validation of direct detection of foot-and-mouth disease virus within clinical samples using reverse transcription loop-mediated isothermal amplification coupled with a simple lateral flow device for detection. PLoS One 9:e105630PubMedPubMed CentralView ArticleGoogle Scholar
- Friedman CR, Neimann J, Wegener HC, Tauxe RV (2000) Epidemiology of Campylobacter jejuni infections in the United States and other industrialized nations. Campylobacter, 2nd edn. ASM International, Washington, pp 121–138Google Scholar
- Lin J (2009) Novel approaches for Campylobacter control in poultry. Foodborne Pathog Dis 6:755–765PubMedPubMed CentralView ArticleGoogle Scholar
- Buzby JC, Allos BM, Roberts T (1997) The economic burden of Campylobacter-associated Guillain-Barré syndrome. J Infect Dis 176:S192–S197PubMedView ArticleGoogle Scholar
- Buzby JC, Roberts T (1996) Economic costs and trade impacts of microbial foodborne illness. World Health Stat Q 50:57–66Google Scholar
- Wangroongsarb P, Jittaprasatsin C, Suwannasing S, Suthivarakom K, Khamthalang T (2011) Identification of genus Campylobacter and four enteropathogenic Campylobacter species by PCR. J Trop Med Parasitol 34:17–29Google Scholar
- Barletta F, Mercado E, Lluque A, Ruiz J, Cleary T, Ochoa T (2013) Multiplex real-time PCR for detection of Campylobacter, Salmonella, and Shigella. J Clin Microbiol 51:2822–2829PubMedPubMed CentralView ArticleGoogle Scholar
- Cecchini F, Manzano M, Mandabi Y, Perelman E, Marks RS (2012) Chemiluminescent DNA optical fibre sensor for Brettanomyces bruxellensis detection. J Biotechnol 157:25–30PubMedView ArticleGoogle Scholar
- Jia K, Toury T, Ionescu RE (2012) Fabrication of an atrazine acoustic immunosensor based on a drop-deposition procedure. IEEE Trans Ultrason Ferroelectr Freq Control 59:2015–2021PubMedGoogle Scholar
- Kubičárová T, Fojta M, Vidic J, Tomschik M, Suznjevic D, Paleček E (2000) Voltammetric and chronopotentiometric measurements with nucleic acid-modified mercury film on a glassy carbon electrode. Electroanalysis 12:1390–1396View ArticleGoogle Scholar
- Lu L, Jun S (2012) Evaluation of a microwire sensor functionalized to detect Escherichia coli bacterial cells. Biosens Bioelectron 36:257–261PubMedView ArticleGoogle Scholar
- Manzano M, Vizzini P, Jia K, Adam PM, Ionescu RE (2016) Development of localized surface plasmon resonance biosensors for the detection of Brettanomyces bruxellensis in wine. Sens Actuator B Chem 223:295–300View ArticleGoogle Scholar
- Manzano M, Cecchini F, Fontanot M, Iacumin L, Comi G, Melpignano P (2015) OLED-based DNA biochip for Campylobacter spp. detection in poultry meat samples. Biosens Bioelectron 66:271–276PubMedView ArticleGoogle Scholar
- Zhang L, Wang O, Cheng ZY, Huang TS (2015) Silica-coated magnetostrictive biosensors for real-time detection of Campylobacter jejuni in washing water. In: Meeting abstracts, The Electrochemical Society, Pennington, pp 1832Google Scholar
- Wei D, Oyarzabal OA, Huang TS, Balasubramanian S, Sista S, Simonian AL (2007) Development of a surface plasmon resonance biosensor for the identification of Campylobacter jejuni. J Microbiol Methods 69:78–85PubMedView ArticleGoogle Scholar
- Gnanaprakasa TJ, Oyarzabal OA, Olsen EV, Pedrosa VA, Simonian AL (2011) Tethered DNA scaffolds on optical sensor platforms for detection of hipO gene from Campylobacter jejuni. Sens Actuators B Chem 156:304–311View ArticleGoogle Scholar
- Wadl M, Pölzler T, Flekna G, Thompson L, Slaghuis J, Köfer J, Hein I, Wagner M (2009) Easy-to-use rapid test for direct detection of Campylobacter spp. in chicken feces. J Food Prot 72:2483–2488PubMedView ArticleGoogle Scholar
- Alcaine S, Soyer Y, Warnick L, Su W, Sukhnanand S, Richards J, Fortes E (2006) Multilocus 519 sequence typing supports the hypothesis that cow-and human-associated Salmonella 520 isolates represent distinct and overlapping populations. Appl Environ Microbiol 521:7575–7585View ArticleGoogle Scholar
- Jasson V, Jacxsens L, Luning P, Rajkovic A, Uyttendaele M (2010) Alternative microbial methods: an overview and selection criteria. Food Microbiol 27:710–730PubMedView ArticleGoogle Scholar
- Lee KM, Runyon M, Herrman TJ, Phillips R, Hsieh J (2015) Review of Salmonella detection and identification methods: aspects of rapid emergency response and food safety. Food Control 47:264–276View ArticleGoogle Scholar
- Chai Y, Li S, Horikawa S, Park MK, Vodyanoy V, Chin BA (2012) Rapid and sensitive detection of Salmonella typhimurium on eggshells by using wireless biosensors. J Food Prot 75:631–636PubMedView ArticleGoogle Scholar
- Bulut O (2014) Development of nucleic acid based lateral flow immunochromatographic test platform for salmonella detection. Ph.D. Thesis Middle East Technical UniversityGoogle Scholar
- Fang Z, Wu W, Lu X, Zeng L (2014) Lateral flow biosensor for DNA extraction-free detection of salmonella based on aptamer mediated strand displacement amplification. Biosens Bioelectron 56:192–197PubMedView ArticleGoogle Scholar
- Kim G, Moon JH, Moh CY, Lim JG (2015) A microfluidic nano-biosensor for the detection of pathogenic Salmonella. Biosens Bioelectron 67:243–247PubMedView ArticleGoogle Scholar
- Cai Z, Song Y, Wu Y, Zhu Z, Yang CJ, Chen X (2013) An electrochemical sensor based on label-free functional allosteric molecular beacons for detection target DNA/miRNA. Biosens Bioelectron 41:783–788PubMedView ArticleGoogle Scholar
- Rochelet M, Solanas S, Grossiord C, Maréchal P, Résa C, Vienney F, Barranger C, Joannes M (2012) A thin layer-based amperometric enzyme immunoassay for the rapid and sensitive diagnosis of respiratory syncytial virus infections. Talanta 100:139–144PubMedView ArticleGoogle Scholar
- Perez JW, Vargis EA, Russ PK, Haselton FR, Wright DW (2011) Detection of respiratory syncytial virus using nanoparticle amplified immuno-polymerase chain reaction. Anal Biochem 410:141–148PubMedView ArticleGoogle Scholar